NRG S1 Anemometer | Initial Setup
- Lay out cables and required tools:NRG Flat Head Screw DriverHex Driver1/2" Open End Wrench
- Remove sensor from box.
- Pull cable-end through boom. Attach to sensor and tighten with wrench. Mount sensor on boom and tighten with hex driver.
- Connect sensor to SymphoniePRO wiring panel. Use channels 1-12.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
110S Temperature Sensor
200M Wind Vane
200P Wind Vane
BP20 Barometric Pressure Sensor
BP60 + BP60C Barometric Pressure Sensor
HybridMC Anemometer + Vane
HybridXT Anemometer + Vane
PVT1 PV Temperature Sensor
T60 + T60C Temperature Sensor
RH5X Relative Humidity Sensor
Thies First Class Advanced Anemometer
WindSensor P2546-OPR Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.