A 110S temperature sensor connected to a 9200-PLUS NRG Logger’s Extra Analog Channel is showing high readings such as 214
The 9200-Plus logger's extra analog channel is set up to read every 10mV of input as one count". The logger is measuring the voltage output of whatever sensor is connected. The ASCII data file will show counts, which could range from 0 to a max of 255 counts. The 110S sensor output will provide 10mV per each degree F of temperature with an offset of 123.5. Here is how to calculate your temperature: Degrees F=((counts x .01V)-1.235V)/.01 For example: ASCII data reading of 214 counts ((214 x .01V)-1.235V)/.01V = 90.5 degrees F.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
110S Temperature Sensor
200M Wind Vane
200P Wind Vane
BP20 Barometric Pressure Sensor
PVT1 PV Temperature Sensor
60 + T60C Temperature Sensor
Thies First Class Advanced Anemometer
WindSensor P2546-OPR Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.