How have wind direction / wind vane measurements changed with SymphoniePRO?
- The deadband can be specified for any connected wind direction vane. For example, if a vane has deadband of 4 degrees centered on the vane “North” mark, this information can be entered as 2 East and 2 West, and the logger will scale the measurements accordingly.
- Vanes with no deadband that short at the North mark can be connected to SymphoniePRO without any special settings required.
- A new parameter called the “vane mounting angle” has been added which accounts for the vane “north” position relative to the boom it is mounted on. The wind direction is calculated accounting for both the boom orientation and the vane mounting angle. Default settings assume the vane north mark is pointed at the tower as the majority of users tend to use this configuration. If the vane’s north mark is not pointed at the tower, the vane mounting angle will need to be adjusted by the user accordingly.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
110S Temperature Sensor
200M Wind Vane
200P Wind Vane
BP20 Barometric Pressure Sensor
PVT1 PV Temperature Sensor
60 + T60C Temperature Sensor
Thies First Class Advanced Anemometer
WindSensor P2546-OPR Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.