Testing RH5X Relative Humidity Sensor
The RH5X sensor uses a polymer resistor sensor to measure relative humidity. By checking the voltage output and current draw of the sensor its health can be determined. The RH5X will run on excitation voltage from 10 to 30 V DC, and will draw approximately 3mA of current.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
BP20 Barometric Pressure Sensor
Windsensor P2546-OPR Anemometers
NRG Class 1 Anemometer
Thies First Class Advanced Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.