Post Calibrating the 40C Anemometer in the SOH Wind Engineering (SOH WE) Wind Tunnel
Published on October 8, 2013
Through October 2012, all 40C non-MEASNET initial production calibrations have been carried out in the Otech Engineering wind tunnels. Some users of 40C Anemometers have had calibrations performed on these anemometers after field-deployment as a quality check. The calibrations performed after field-deployment are often referred to as “post-calibrations” and are typically performed in the same wind tunnel as the initial calibration (to remove tunnel-to-tunnel bias in the result).
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
110S Temperature Sensor
200M Wind Vane
200P Wind Vane
BP20 Barometric Pressure Sensor
PVT1 PV Temperature Sensor
60 + T60C Temperature Sensor
Thies First Class Advanced Anemometer
WindSensor P2546-OPR Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.