Wind speed-readings are zero
If wind-speed readings are zero:
- Check all the sensor connections. Be sure that the sensor is connected with the proper polarity.
- Check that all wires are intact. Resistance across the anemometer should be approximately 600 to 700 ohms (if not spinning). A spinning anemometer will read approximately 1 - 2 volts peak-to-peak AC at 25 mph.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
110S Temperature Sensor
200M Wind Vane
200P Wind Vane
BP20 Barometric Pressure Sensor
PVT1 PV Temperature Sensor
60 + T60C Temperature Sensor
Thies First Class Advanced Anemometer
WindSensor P2546-OPR Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.