What is the class of the NRG Class 1 Anemometer?
The IEC Standard recommends certain cup classifications for anemometers to be used in power performance measurements (not wind assessment). Specifically, Annex I of IEC standard 61400-12-1 defines the general influence parameter ranges to use in the assessment of anemometer response deviations. The resultant maximum anemometer response deviation is the class number k.
Based on work performed by RNRG and endorsed by Troels Pedersen of DTU Wind Energy Department (read White Paper), the classification indices for the RNRG Class 1 anemometer are 1.01A for Type A terrain and 8.44B for Type B terrain.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
BP20 Barometric Pressure Sensor
Windsensor P2546-OPR Anemometers
NRG Class 1 Anemometer
Thies First Class Advanced Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.