What are the main differences between the Class 1 Anemometer and the 40C Anemometer?
Primary differences between the two sensors are threshold, inertia, and distance constant. In order to obtain Class 1 compliance, the Rulon bearings currently found in the 40C Anemometer have been replaced with a new design that includes a ball-bearing cartridge. The new patent pending ball-bearing design provides comparable variance and consistency in data as compared with other leading Class 1 Anemometers, at a much more economical price. In addition to being an important element towards meeting Class 1 compliance, the design also offers exceptional dust protection and strength.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
110S Temperature Sensor
200M Wind Vane
200P Wind Vane
BP20 Barometric Pressure Sensor
PVT1 PV Temperature Sensor
60 + T60C Temperature Sensor
Thies First Class Advanced Anemometer
WindSensor P2546-OPR Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.