Using a long cable with a 200P wind vane
We have tested the #200P with cables up to 1525m (5000 ft.). While there is no degradation of signal strength or sensor accuracy, longer cables always increase the risk of electrical noise.
Splice points should be carefully covered in heat shrink tubing or high quality electrical tape appropriate for outdoor use. In electrically noisy environments, long cables should be avoided.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
BP20 Barometric Pressure Sensor
Windsensor P2546-OPR Anemometers
NRG Class 1 Anemometer
Thies First Class Advanced Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.