USB to Serial Converters
If you are using a serial to USB converter to connect through the logger programming port, there could be a problem with the USB device configuration. Sometimes, shutting down the SDR "Modify iPack Settings" window is enough to restart the USB to Serial device. If the USB to Serial converter is brand new, it may require the installation of drivers in order to operate. Further, you may need to investigate your hardware profile in the Windows Device Manager control panel and make sure the device is properly activated and is mapped to a COM port 1-4.
If all else fails, you can restart your computer.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
110S Temperature Sensor
200M Wind Vane
200P Wind Vane
BP20 Barometric Pressure Sensor
PVT1 PV Temperature Sensor
60 + T60C Temperature Sensor
Thies First Class Advanced Anemometer
WindSensor P2546-OPR Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.