SDR informs me that my logger settings may have changed...how does it know?
The Symphonie logger will set a configuration change "flag bit" which is passed to SDR through the rwd file header.
The flag bit can be set under the following conditions:
2.The Encryption PIN is changed
3.A SCM change is detected
The logger may have experienced a mild ESD, which caused the reboot and no components are damaged or in the event of a very strong ESD to the logger (nearby lightning) the SCM identification circuits and signal inputs may sustain damage. It is good practice to inspect any files in which SDR has prompted you to review changes.
Note that the SCM configuration is scanned under the following conditions:
1.Every two seconds that the logger user interface (display) is active
2.At the beginning of every 10 minute interval
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
110S Temperature Sensor
200M Wind Vane
200P Wind Vane
BP20 Barometric Pressure Sensor
PVT1 PV Temperature Sensor
60 + T60C Temperature Sensor
Thies First Class Advanced Anemometer
WindSensor P2546-OPR Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.