Recommended Cleaning Method for Pyranometers
We recommend the following method for cleaning a pyranometer dome:
Step 1: Use a spray bottle filled with distilled water to vigorously spray the dome to wash away any particulates from the dome surface.
Step 2: If there is any hardened particulate stuck to the dome surface (e.g. bird droppings), spray with distilled water and then cover with a moist/damp paper towel for five minutes to loosen the hardened particulate, then spray vigorously again with distilled water to wash away the contaminant.
Step 3: Use a clean lens wipe to remove any remaining contaminants on the dome surface. Note: Discard the lens wipe after each use. Do not use it to clean another dome (one lens wipe per dome).
Step 4: This step is optional, but if you have a clean microfiber cloth, you can do a final polish of the dome once it has been cleaned with the lens wipe in Step 3.
*Cleaning recommendations courtesy of Robert Dolce, Managing Director, HuksefluxUSA Inc.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
BP20 Barometric Pressure Sensor
Windsensor P2546-OPR Anemometers
NRG Class 1 Anemometer
Thies First Class Advanced Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.