Installing NRG sensors on a lattice tower
There are a variety of opinions regarding the recommended distance between the sensors and tower, but three times the lattice tower face width seems to be a popular guideline.
However, in cases where that's not possible, we recommend putting the sensor at least 1-2 face widths directly into the prevailing wind direction (or at least downwind from the direction of the fewest winds).
Most Rohn lattice towers have a face width of 18 - 24 inches (46 - 61 cm), so mounting booms need to keep the sensors at least 48 - 72 inches (1.2 - 1.8 m) from the nearest tower leg.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
110S Temperature Sensor
200M Wind Vane
200P Wind Vane
BP20 Barometric Pressure Sensor
PVT1 PV Temperature Sensor
60 + T60C Temperature Sensor
Thies First Class Advanced Anemometer
WindSensor P2546-OPR Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.