For interpreting 40C stored signal data, should I continue to use NRG’s published 40 Consensus Transfer Function (CTF), or should I use the specific wind tunnel, calibration test-derived value (value provided on calibration facility’s test certificate)?
You can use either. Use of the consensus transfer function remains a good choice. It is also acceptable to use the individual wind tunnel measured values provided with the sensor.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
110S Temperature Sensor
200M Wind Vane
200P Wind Vane
BP20 Barometric Pressure Sensor
PVT1 PV Temperature Sensor
60 + T60C Temperature Sensor
Thies First Class Advanced Anemometer
WindSensor P2546-OPR Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.