Extending the cable for a 110S temperature sensor
The cable may be extended up to approximately 500 meters before the excitation signal will degrade and affect the operation of the temperature sensor. Cable should be 20 gauge and MUST be shielded. It is recommended that splices are carefully wrapped with high quality electrical tape or covered with heat shrink tubing appropriate for outdoor use. Connect shields really well to prevent electrical noise.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
BP20 Barometric Pressure Sensor
Windsensor P2546-OPR Anemometers
NRG Class 1 Anemometer
Thies First Class Advanced Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.