Does a WindSensor P2546A anemometer require an iPack to work? How do I wire a WindSensor P2546A?
When connecting a WindSensor P2546A anemometer to the SymphoniePLUS3’s preconfigured channels (1-3, 13-15) an iPack is required. It should be wired:
- Brown to Excitation
- White to Signal
This is because these channels use the 12V excitation from the iPack to create pulses that get fed into the channel as the sensor’s Reed switch turns on and off.
When using the SymphoniePLUS3’s flex channels (4-6) with a Digital Anemometer SCM no iPack is required.
This configuration it should be wired:
- Brown to Signal
- White to Ground
When using a Symphonie or SymphoniePLUS logger, counter channels (4-6) with a Digital Anemometer SCM, no
iPack is required. This configuration should be wired:
- Brown to (-)
- White to (+)
The Digital Anemometer SCM routes the data logger’s 5V through the sensor’s Reed switch.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
110S Temperature Sensor
200M Wind Vane
200P Wind Vane
BP20 Barometric Pressure Sensor
PVT1 PV Temperature Sensor
60 + T60C Temperature Sensor
Thies First Class Advanced Anemometer
WindSensor P2546-OPR Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.