WindSensor Cable Termination
These instructions illustrate how to terminate a male LEMO Push-Pull connector to 2 conductor cable for use with a WindSensor P2546-OPR anemometer (either "A" or "C" model). Termination of this connector requires precise cable stripping and soldering equipment. When properly terminated, the resulting connection should provide reliable, weatherproof performance for the duration of the measurement campaign.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
110S Temperature Sensor
200M Wind Vane
200P Wind Vane
BP20 Barometric Pressure Sensor
PVT1 PV Temperature Sensor
60 + T60C Temperature Sensor
Thies First Class Advanced Anemometer
WindSensor P2546-OPR Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.