Using Thermopile Pyranometers on Symphonie Loggers
Symphonie loggers now support thermopile pyranometers using a choice of two optional Signal Conditioning Modules (SCMs). Thermopile pyranometers provide a signal output proportional to the total solar radiation, measured in Watts per square meter (W/m2 ). Each pyranometer includes a calibration sheet which specifies the sensitivity for that individual sensor. The logger collects and stores data in raw format; and a scale factor needs to be calculated to convert the raw data to engineering units.
This application note explains how to calculate the correct scale factor based on your individual pyranometer calibration value (sensitivity).
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
110S Temperature Sensor
200M Wind Vane
200P Wind Vane
BP20 Barometric Pressure Sensor
PVT1 PV Temperature Sensor
60 + T60C Temperature Sensor
Thies First Class Advanced Anemometer
WindSensor P2546-OPR Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.