Replacing iPack Batteries
Lead acid batteries such as the one used in the Symphonie iPack need to be kept fully charged. Battery capacity will be reduced if the battery is allowed to discharge below 10.5 volts for an extended period of time. If your iPack battery has been completely discharged (due to PV not charging or connected improperly, for example) it may need to be replaced. This is especially true in colder environments where battery capacity is further reduced due to the low temperatures.
This application note explains how to change an iPack battery.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
110S Temperature Sensor
200M Wind Vane
200P Wind Vane
BP20 Barometric Pressure Sensor
PVT1 PV Temperature Sensor
60 + T60C Temperature Sensor
Thies First Class Advanced Anemometer
WindSensor P2546-OPR Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.