Connecting a Windsensor (RISO) P2546A Anemometer to Symphonie Loggers
The Windsensor P2546A anemometer (commonly referred to as the RISO anemometer) is fully compatible with all Symphonie loggers.
For Symphonie and SymphoniePLUS loggers:
- Use channels 4, 5 or 6
- Use 3151 Digital Anemometer SCM (see product page)
The connections are as follows
- WHITE wire to Ctr +
- BROWN wire to Ctr -
All P2546A anemometers are individually calibrated and are shipped with calibration slope and offset parameters. In lieu of using those values, the following slope and offset are commonly used:
- Slope (scale factor) = 0.6201 m/s
- Offset = 0.270 m/s
SymphoniePLUS3 loggers can use the above settings, or can accommodate the sensor in any of the preconfigured anemometer channels; 1, 2, 3, 13, 14, and 15.
Download Calibration Reports
Our Calibration Report Retriever Tool is a comprehensive database of reports for the following products:
110S Temperature Sensor
200M Wind Vane
200P Wind Vane
BP20 Barometric Pressure Sensor
PVT1 PV Temperature Sensor
60 + T60C Temperature Sensor
Thies First Class Advanced Anemometer
WindSensor P2546-OPR Anemometer
Whether you are troubleshooting in the field or learning how to install a product, we are here for you.